Each DSCL23 loop powered isolator provides 1 or 2 channels of analog signal protection by electrically isolating its input from output for any DC process signal in the range 4-20mA. The DSCL23 feeds isolated DC power to the field transmitter and returns a fully isolated 4-20mA analog signal.
The input signal is chopped by a proprietary DC chopper circuit, isolation is provided by transformer coupling, a proprietary technique is used to suppress transmission of common mode spikes or surges, and the output signal from the secondary of the transformer is reconstructed to the original input signal.
Frequently Asked Questions
Can Dataforth provide calibration reports for modules I purchased?
Yes we can provide calibration reports for the modules that you purchased.
You can either
a) visit https://www.dataforth.com/TestDataReport.aspx to search for Test Report Datasheets by Serial Number or
b) you can send us a list of model numbers and their serial numbers to firstname.lastname@example.org
Does Dataforth have a 4-channel 4-20mA isolator?
No. However, Dataforth does have two and three-channel 4-20mA isolators as shown in the following table.
DSCL22-11 2-channel 4-20mA isolator
DSCL22-21 3-channel 4-20mA isolator
DSCL23-02 2-channel 4-20mA Isolator with DC Supply for 2-wire transmitter
DSCL24-11 2-channel 4-20mA isolator
Keywords/Phrases: 4-20mA isolator, 4-channel 4-20mA isolator, 4-channel 4-20mA DIN mount isolator
Why do I see a -10V shift when I change to a DSCL23 from a DSCA42-01C or similar?
The DSCA42-01C internal electronics is powered by an overall power supply and then the module output circuitry provides the output current whose return side (or negative side) is connected internally to the power supply common. This causes the output signal to be positive with respect to the common.
Whereas, the DSCL23 is loop-powered through its output terminals. That puts the 4-20mA receiver (the load) in series with the power supply. So when you measure the voltage from the power supply common (- terminal) to the other side of the load you will see a negative signal. The module could be made usable if the power supply has a fully floating output and you can make terminal 1 (the return current terminal) the common and measure the positive signal across the load from the power supply (-) to terminal 1 (the return current terminal), the common.
A better match for your application is the DSCA42-01C. Its internal electronics is powered by an overall power supply and then the module output circuitry provides the output current whose return side (or negative side) can be connected to the power supply common. This causes the output signal to be positive with respect to the common.
Keywords/Phrases: 4-20mA isolator with loop power, 2-wire transmitter interface signal conditioner with loop power
How does the load resistance of a module affect the noise at the output?
Noise at the output of a module is independent of load resistance.
Was this content helpful?
View Dataforth cross-reference data.
- No cross reference is available.
The information available through this competitive cross reference guide are based upon product catalog information obtained from a variety of sources. The competitive cross reference information is being provided to you free of charge for your use. While Dataforth Corp has used reasonable efforts to ensure data accuracy, Dataforth Corp does not guarantee that it is error-free, nor does Dataforth Corp make any other representation, warranty or guarantee that the information is accurate, correct, reliable or up-to-date. Dataforth Corp expressly disclaims all implied warranties regarding this information, including but not limited to any implied warranties of merchantability or fitness for a particular purpose.
This information is provided only as a convenience on an "as is" basis and Dataforth Corp. or its representatives or distributors are not responsible for any incorrect, inaccurate, or incomplete information. You are solely responsible for (1) selecting the appropriate Dataforth products for your application, (2) designing, validating and testing your application, and (3) ensuring your application meets applicable standards, and any other safety, security, or other requirements.